Dyne Testing is a technology, a method to measure surface wettability. The low surface wettability of polymer-based substrates is the sign of poor adhesion of inks, glues and coatings. Thus, to obtain the optimum amount of adhesive it is necessary to increase the surface energy of the substrate which can be done by surface treatment with either Corona or Plasma. It will result in good wetting of the material over the surface of the substrate and hence, it improves adhesion.
For the optimum adhesion, while printing, gluing, or coating the various substrates, it is necessary to obtain high surface energy which can be obtained by Dyne Testing Markers. The fluid that is present in the Dyne Testing markers is based on ISO 8296 method for measuring the surface energy of polythene film.
When the Dyne Testing Pens are being applied to the surface, the liquid will form a continues film or will form a small trail of droplets. If it is being stretched as a film for at least 3 seconds, the substrate will have a minimum surface energy of that ink value which will be expressed in mN/m (Dynes).
The exact surface energy (Dyne level) can be determined by applying a range of increasing or decreasing values of Dyne test pens thereby taking the steps to improve its condition.
The Dyne Test Pen may lose its accuracy for which there are 3 reasons:
- It could get contaminated with the foreign substance
- It could evaporate quicker than it is expected to be
- And the third reason is ageing, during which chemical reactions take place among the constituents.
The experts have also faced the problem with the ageing of Dyne Testing Equipment. If their hue or color density are almost past their expiration date, it is advisable to replace them as stated by experts. The lower Dyne value states that the value stated on the bottle does not match true surface tension. You must be sure that retains the substrate used for the measurements are to be kept well sealed, free from contamination, and stored under laboratory conditions.
The ideal Dyne Testing Pen should be:
- easy to handle,
- perfect for the quick spot checks on the production floor,
- very easy to read,
- no subjectivity for this type of test,
- no wiping off necessary,
- lasting display of result, and
- very striking coloring.
The Dyne Testing Kit by Ice-equipment.com is based on valve tip applicator and not the magic marker type. The quick test 38 pen is our most popular product amongst all and it is available in a bright red ink. This is the quick test pen which serves to check the surface treatment of all plastic substrates.
It has a shown an effect onto the material such that a stroke of the pens leaves a full line on the material if the material’s surface energy is below 38 Dynes/cm. Also, as mentioned above if the materials surface energy is below 38 Dynes/cm, the fluid will form small drops on the surface. The fluid applied to the surface will dry within seconds; it does not need to be wiped off anymore.